« السابقةمتابعة »
ARTS, SCIENCES, AND MISCELLANEOUS
ENLARGED AND IMPROVED.
THE SIXTH EDITION.
Illustrated with nearly sir hundred Engravings.
INDOCTI DISCANT; AMENT MEMINISSE PERITI.
AND HURST, ROBINSON, AND COMPANY, 90, CHEAPSIDE,
GENI LEO HIST. ANTIQ. WALFORD 10-8-68 711776-129 ADD VOLI
M I C
M I C
the plates in a right position and counteract the long scupe. bapa objects appear larger than they do ta the naked eye. screw CC. I is a small turned handle, for the better
Single microscopes consist of a single lens or mirror; or holding of the instrument, to screw on or off at plea-
the seventh magnifier, set in the manner of a little For the principles on which the construction of mi- barrel, to be beld in the land for the viewing of any croseopes depends, see Optics. In the present article, larger object. M is a flat slip of ivory, called a it is intended to describe the finished instrument, with slider, with four round holes through it, wherein to all its varied apparatus, according to the latest improve place objects between two pieces of glass or Muscovy ments ; and to illastrate by proper details its uses and talc, as they appear at ddd d. Six such sliders, and importance.
one of brass, are usually sold with this mioroscope,
some with objects placed in them, and others empty I. Of SINGLE Microscopes.
for viewing any thing that may offer : but whoever
pleases to make a collection, may bave as many as he The famous microscopes made use of by Mr Leeo- desires. The brass slider is to confine any small'ubject, wenhoeck, were all, as Mr Baker assures us, of the thet it may be viewed without crushing or destroying single kind, and the construction of them was the most it. N is a tube of glass contrived to confine living ob simple possible, each consisting only of a single lens set jects, such as frogs, fishes, &c. in order to discover the between two plates of silver, perforated with a small circulation of the blood. All these are contained in a hole, with a moveable pin before it to place the object little neat box of fish-skin or mahogany, very convenient on and adjust it to the eye of the beholder. He informs for carrying in the pocket. us also, that lenses only, and not globules, were used in Wben an -object is to be viewed, thrust the ivory every one of these microscopes.
slider, in which the said object is placed, between the 1. The single microscope now most generally known two flat brass plates EE: observing always to put CECXXIVII, and used is that called Wilson's Pocket Microscope. The that side of the slider where the brass rings are farbody is made of brass, ivory, or silver, and is repre
thest from the eye. Then screw on the magnifying sented by A4, BB.CC is a long fine threaded male glass you intend to use, at the end of the instrument screw that turns into the body of the microscope ; D G; and looking through it agaiuse the light, turn the & convex glass at the end of the screw. Two con- long screw CC, till your object be brought to suit your cave round pieces of thin brass, with holes of different eye; which will be known by its appearing perfectly diameters in the middle of them, are placed to cover
distinct and clear. It is most proper to look at it first the above-mentioned glass, and thereby diminish the through a magnifier that can show the whole at once, aperture when the greatest magnifiers are employed. and afterwards to inspect the several parts more partiEE, three this plates of brass within the body of the cularly with one of the greatest magnifiers; for thus microscope ; one of which is bent semicircularly in the you will gain a true idea
of the whole, and of all its middle, so as to form an arebed cavity for the recep parts. And though the greatest magnifiers can show tion of a tube of glass, the use of the other two be but a minute portion of any object at once, such as the ing to receive and hold the sliders between them. F, claw of a flea, the horn of a louse, or the like ; yet by a piece of wood or ivory, arched in the manner of gently moving the slider which contains the object, the the semicircular plate, and cemented to it. G, the eye may gradually examine it all over. other end of the body of the microscope, where a hol. As objects must be brought very near the glasses low female screw is adapted to receive the different when the greatest magnifiers are made use of, be caremagnifiers. H is a spiral spring of steel, between ful not to scratch them by rubbing the slider against Vol. XIV. Part I. +
· Fig. 1.